HAMEG INSTRUMENTS HM | Module: function generator; 7-Segment,5- digit LED; Band: ≤10MHz – This product is available in Transfer Multisort. F requency Range 50 mHz 10 MHz,. Output Voltage up to 10 Vpp (into 50 Ω). ☑ Waveforms: Sine Wave, Triangle, Square Wave, Pulse, DC. ☑ D istortion Factor. Funktionsgenerator HM Equipment HAMEG GmbH, Frankfurt, build ?, 4 pictures, 4 schematics, Germany, tubes, semiconductors, Service- or Lab.
|Published (Last):||1 March 2008|
|PDF File Size:||16.54 Mb|
|ePub File Size:||3.45 Mb|
|Price:||Free* [*Free Regsitration Required]|
Its low frequency ranges are particularly well suited for simulating mechanical and servo techniques. Do not appiy any DC voltage to the out- put socket. Connect the distortion meter to TP1.
HM8030-5 function generator module,5MHz
The selected frequency appears on the 4-digit display 0. Compared to knob scales, this display has a much higher resolution. Adjust signal height to 6 div.
You agree that the Information as provided here by RS may not be error-free, accurate or up-to-date and that it is not advice. All case and chassis parts are connected to the safety earth conductor. H 68, D mm Weight: An offset voltage adjusted at the 50 ohms output has no influence upon the hameb signal.
DC offset If the offset knob is not in its calibrated position, a DC voltage can be superimposed on hamsg output signal.
The Nameg and kHz range indicators are integrated into the dis- play panel. The frequency ranges are adjusted in the following order: Warranty Before being shipped, each plug-in module must pass a 10 hour quality control test.
HM Function Generator Module | Overview | Rohde & Schwarz
The set frequency Nq DC voltage not included can be selected at will. Safety Every module is manufactured and tested for use only with the mainframe HM according to lEC Part 1 and la Safety requirements for electronic test and measurement equipment. Trigger output In the sine, square and triangle modes, the trigger out- put 0 supplies a square signal in synchronism with the output signal. Save to parts list Save to parts list.
The opera- tional position is optional, however, the ventilation holes on the HM and on the plug-in modules must not be obstructed.
Therefore only short or low-capacitance cables without a 50 ohms ter- minating resistor should be used for connection.
The required trimmer is located on the upper PCB next to the amplitude control.
Providing the finest test equipment solutions since 1992
Set the trigger haeg con- trol slope of the oscilloscope to negative edge trigger- ing. If exact square-shaped signals are hmaeg, care should be taken that only 50 ohms coaxial cables e.
Adjustment procedure The following instructions will permit you to correct any deviation of the module from the nominal data. The TP 1 test point is located next to the amplitude control on the upper PC board. Pull out the previously used module by its handle and slide in the new module until the end position is reached.
Sending feedback, please wait They can be used separately. When removing or replacing the metal case, the instru- ment must be completely disconnected from the mains supply. Additional quality fea- tures include the relatively low distortion factor of the generated signals and constant amplitude flatness throughout the entire frequency range of the instrument. Connect multimeter to TP1.
In hhameg signal volt- age ranges, it should be noted that the used terminating resistor must dissipate the corresponding effective power.
A negative DC voltage reduces the frequency. Before exchanging the module, the mainframe must be switched off. We regret that transportation damage due to poor packaging is not covered by this warranty.